|[Date Prev][Date Next][Thread Prev][Thread Next][Date Index][Thread Index]|
BagEasy Manual Resuscitation Devices by Westmed, Inc.: Class 1 Recall
AUDIENCE: Risk Manager, Emergency Medicine
ISSUE: Westmed, Inc. is initiating a nationwide recall of 24,384 units of BagEasy Manual Resuscitation Devices. The BagEasy device have been found to have a potential for disconnection at the retention ring of the patient port manifold. Disconnection causes the unit to be inoperable, which potentially could result in treatment delays while another unit is obtained or technician switches to a different method of resuscitation.
BACKGROUND: Westmed has become aware of a potential for disconnection at the patient port retention ring assembly of the BagEasy manual resuscitation device after receiving reports from three separate facilities regarding units disconnecting during setup or use. The BagEasy manual resuscitation device was distributed to other medical device distribution companies as well as directly to hospitals and other treatment facilities.
RECOMMENDATIONS: Consumers with questions may contact the company at 1-800-975-7987 and arrange for return of all recalled products. For the affected part numbers and lot numbers please see the firm press release. Healthcare professionals are encouraged to report adverse events or side effects related to the use of these products to the FDA's MedWatch Safety Information and Adverse Event Reporting Program:
Read the complete MedWatch Safety Alert at: http://www.fda.gov/Safety/MedWatch/SafetyInformation/SafetyAlertsforHumanMedicalProducts/ucm226014.htm
Tell us what you think: http://survey.foreseeresults.com/survey/display?cid=d509U9xg0gZVxkMRcoMsQg==&sid=link&cpp[date]=09_15_2010_2000&cpp[type]=S
You are encouraged to report all serious adverse events and product quality problems to FDA MedWatch at www.fda.gov/medwatch/report.htm
Update your subscriptions, modify your e-mail address, or stop subscriptions at any time on your Subscriber Preferences Page. You will need to use your e-mail address to log in. If you have questions or problems with the subscription service, please contact support@xxxxxxxxxxxxxxx.
This service is provided to you at no charge by U.S. Food & Drug Administration (FDA).
U.S. Food & Drug Administration (FDA) · 10903 New Hampshire Ave · Silver Spring, MD 20993 · 800-439-1420